# Kinemetrics Episensor2

Nominal Response Library
Element sensor
Manufacturer Kinemetrics
Model Episensor2

### Full-Scale Voltage

Full-Scale Voltage is the range of voltages output by the sensor for the largest and smallest acceleration described by the clip level.To illustrate, if an accelerometer with a +/-4g clip level outputs +10V for a +4g signal and -10V for a -4g signal, the full-scale voltage is +/-10V. Here are three equivalent ways to express this:- the full scale voltage is +/- 10V,- the peak-to-peak voltage (Vpp) is 20V (+10 to -10), and - the peak voltage (Vp) is 10V.

### Output Wiring

Instruments can be wired for either single-ended (one positive) or differential (one positive and one negative) output. In the case of differential output, the instrument reverses the polarity of the negative output before summing it with the postive to amplify signal and cancel induced noise. For an instrument that can be run in either mode, the full-scale voltage in differential mode will be twice the full-scale voltage in single-ended mode.

### Sensitivity

The nominal sensitivity for an accelerometer in V/m/s**2 is

[Full-scale voltage / Clip level] / [9.80665 m/(g * s**2)]

Sensitivities for clip and full-scale voltage values not listed here can be calculated using this relationship and substituted.

## Configurations

Updating...

Showing 5 of 5

Clip Level
Full-Scale Voltage
Output Wiring
High-Frequency Corner
Sensitivity
Sensor Type
Formats
sensor_Kinemetrics_Episensor2_CG0.25g_FV20Vp_OWDifferential_HF200_SG8.1577_STgroundAccel
0.25g 20Vp Differential 200 Hz 8.1577 V/m/s**2 groundAccel
sensor_Kinemetrics_Episensor2_CG0.5g_FV20Vp_OWDifferential_HF200_SG4.0789_STgroundAccel
0.5g 20Vp Differential 200 Hz 4.0789 V/m/s**2 groundAccel
sensor_Kinemetrics_Episensor2_CG1g_FV20Vp_OWDifferential_HF200_SG2.0394_STgroundAccel
1g 20Vp Differential 200 Hz 2.0394 V/m/s**2 groundAccel
sensor_Kinemetrics_Episensor2_CG2g_FV20Vp_OWDifferential_HF200_SG1.0197_STgroundAccel
2g 20Vp Differential 200 Hz 1.0197 V/m/s**2 groundAccel
sensor_Kinemetrics_Episensor2_CG4g_FV20Vp_OWDifferential_HF200_SG0.5099_STgroundAccel
4g 20Vp Differential 200 Hz 0.5099 V/m/s**2 groundAccel